|Time||15:43:00 Local||6th December 2013||21:43:00 UTC|
|Temperature||34.7 °F||2 °C|
|Dew Point||26.9 °F||-3 °C|
|Humidity||73 %||73 %|
|Pressure||30.22 in/Hg||1023.4 mb|
|Wind Speed||18 mph||29 kmh|
|Solar Radiation||128 W/m2||128 W/m2|
Estimated cloudbase: (if clouds are present): 1700 ft
Forecast: Increasing clouds with little temperature change. Precipitation possible within 24 to 48 hours.
In short: Partially cloudy.
Rainfall Today: 0.04 in
Last 24 hours: 0.04 in
This month: 0.05 in
This year: 37.62 in
Last good rain: Today (0.04 in)
The tilt on the wind sensor has been fixed.... I am hoping it is still pointing in the same direction. I think it is. So at least the wind speed should be accurate again.
We have replaced the failed anemometer transmitter unit, and we now have wind speed readings again!
Moved to new software: WeatherCat - which has much more comprehensive reporting and data export, enabling us to reactivate this website properly. New wind sensor is on order.
Disclaimer: This weather is provided by Jo Hunter for the benefit of those who wish to know.
Weather information from this website should not be used for aviation or marine activities.
Please consult official weather sources for operational purposes.